Level indicators manufacturer by Kaidi86? Ground interference refers to one of the two output ends of the radar level gauge (or compensation line), and its AC voltage to the ground is called the ground interference voltage. This disturbance is also known as longitudinal, in-line mode or in-line state disturbance. Generally speaking, the interference voltage to ground can reach several volts or even more than 100 volts. The above is the interference of the radar level gauge we discussed. When we install or use the radar level gauge, we can also determine whether your use environment will cause these interferences to the level gauge according to the above situation. If these conditions exist, anti-interference measures must be taken to ensure that the level gauge can serve you better. Discover additional information on level gauge manufacturers.
Level Measurement Considerations: Taking advantage of a specific technology’s ability to reliably address the level measurement in either of these vessels, especially the blowdown flash tank, in a plug-and-play type installation and commissioning format is an easy way to ensure optimal performance. This forgoes calibration, external hardware or inputs. Estimates of up to 49 percent of the energy can be recovered through the use of flash steam routed to heat exchangers or the deaerator to preheat boiler makeup water or support the deaeration process, respectively. In addition, better level control technology at the boiler side eliminates energy losses resulting from unnecessary blowdown to prevent carryover conditions.
In addition, some silos in cement plants are very high, such as homogenizing silos of 50cm. It takes time and energy to board high silos to debug radar, so it is recommended to choose HART handheld operators that can be debugged remotely in the central control room. In the central control room, the range and other basic parameters can be set, and the radar echo waveform can be observed, and the waveform can be used for remote diagnosis and debugging, greatly reducing the on-site work intensity of the staff, to avoid the risk of climbing operation. The smart radar level gauge commonly used at present also has a function similar to “driving recorder”, that is, when the material surface mutation occurs on the scene, it can capture the radar echo waveform at that time, which is very useful for debugging the silo under complex conditions.
Rod antenna: generally used in strong corrosive environments, with weak anti-interference ability and small range; Flare antenna: stronger anti-interference ability, suitable for more complex environments. The larger the bell mouth, the more concentrated the energy, and the larger the measuring range; Parabolic antenna: the focusing effect is stronger than that of the bell mouth, the anti-interference ability is the strongest, and the range is the largest.
The electromagnetic wave emitted during measurement can pass through the vacuum, no transmission medium is required, and the anti-interference ability is strong, and it is not affected by temperature, wind, water vapor, water mist, rain, etc. It can be used for liquid level measurement of almost all liquids, and can be continuously measured on-line unattended. Large measurement range, high precision, mature technology and stable performance.
KAIDI level transmitter manufacturer is dedicated in providing complete customized solutions for a wide range of industrial automation process applications – in material level, liquid flow, pressure and temperature. We are constantly developing and innovating, our core vision – “to provide solutions that exceed customers’ expectations. In 2012, the company successfully expanded its operations both locally and internationally, achieving global success and recognition for quality fork type level switch, magnetic level gauge products and services. Read more details on kaidi86.com. OEM Service-can be customized according to the application of customer,such as level gauge,level switch and level indicator.
Working principle: Tube in the liquid level meter has a magnetic float, float design, according to the proportion of medium float magnet steel and container liquid level at the same level, wholesale liquid level gauge in a semiconductor magnetel liquid level gauge sensor LED display scale, and orderly arranged from top to bottom, and LED module corresponding to the magnetic field magnetic sensor, the magnetic semiconductor, triggering the corresponding digital circuit, liquid level above the red LED display, The LED below the liquid level gauge is displayed in green, the red in gas phase, and the green in liquid phase. The red-green junction is the actual liquid level in the container.
Under many operating conditions, ultrasonic level meter and radar level meter are commonly used. Some users are very entangled in the choice of these two level meter and do not know how to choose. Today, let’s talk about the principles and selection principles of these two types of level meter . Principle and selection principle of ultrasonic level meter. Working principle: The ultrasonic pulse probe emits a beam of ultrasonic pulses to the measured medium, and the sound wave is reflected by the liquid surface. The distance between the liquid levels is measured by measuring the time difference between the emission and reflection of the sound waves. Since the ultrasonic level gauge is not a liquid, it can be used to measure corrosive, non-volatile and non-foaming places.